Pulse Oximeter | Cypress Semiconductor
A Pulse Oximeter is a non-invasive device for measuring the percentage of arterial blood (or hemoglobin) that is saturated with oxygen.This device can also measure heart rate.Pulse Oximeters are used in medical, sports training and home appliance applications.Knowing the percentage of hemoglobin that is saturated with oxygen is important to provide anesthesia, to determine the effectiveness of a patient's respiratory system, as well as help to diagnose various illnesses.
The principle of operation of a Pulse Oximeter is based on measuring the absorption of red and infrared light that passes through a patient's finger or ear lobe by utilizing light sensors.Hemoglobin that transports oxygen (oxy-hemoglobin) absorbs infrared wavelength (800-940 nm) of light and hemoglobin that does not transport oxygen (deoxy-hemoglobin), absorbs visible RED wavelength (600-700 nm) of light.Backgrounds such as fluid, tissue and bone are factored out of the measurement by monitoring the steady state of absorption from bone, tissue, venous blood and arterial blood.LEDs are used as the light source and are sequentially pulsed at a fast rate.During a heartbeat, blood volume increases and this AC component of the photodetector's current is used to calculate the absorption of oxy- and deoxy-hemoglobin.
PSoC enables a scalable one chip pulse oximeter solution that flexibly integrates all the basic and advance requirements of a pulse oximeter.PSoC can also enable more advanced features for innovating Pulse Oximeters including USB, Graphics Display, and CapSense buttons.Finally, PSoC can also integrate numerous vital sign functions such as Blood Pressure Monitor, IR Thermometer, and Pulse Oximeter into a single chip solution.
AN52927 demonstrates how easy it is to drive a segment LCD glass using the integrated LCD driver in PSoC 3 and PSoC 5LP
AN57821 introduces basic PCB layout practices to achieve 12- to 20-bit performance for the PSoC 3, PSoC 4, and PSoC 5LP family of devices.
AN58304 provides an overview of the analog routing matrix in PSoC® 3 and PSoC 5LP.
AN58827 discusses how internal trace and switch resistance can affect the performance of a design and how these issues can be avoided by understanding a few basic details about the PSoC® 3 and PSoC 5LP internal analog architecture.
This application note describes how to configure the PSoC® 3 and PSoC 5LP IDACs as a flexible analog source.It presents different approaches for using the IDACs in applications, and discusses the advantages and disadvantages of the topologies presented.This application note will:help you to understand compliance voltage and why it is important; explain how to generate an “any range” or “any ground” VDAC; describe an implementation for a multiplying VDAC; give details on how to build a rail-to-rail low-output impedance 9-bit VDAC from a single IDAC, an opamp, and a resistor; and provide information on how to build a current scaling circuit with an opamp and two resistors.
AN60590 explains diode-based temperature measurement using PSoC® 3, PSoC 4, and PSoC 5LP.